Selecting Create Task in the menu opens a window where you can select the measurement type, see Figure Learn More – opens in a new window or tab Any international shipping is paid in part to Pitney Bowes Inc. We may then repeat this procedure for other channels. Digital input DI and digital output DO: There are 1 items available. We will perform a very simple test, which is sufficient if we are to check that both AO0 and AI0 work correctly.
|Date Added:||8 October 2008|
|File Size:||23.16 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
On-board ni daq 6009 sources available at individual terminals: In the folllowing it described how to create tasks for the analog input and the analog output operations. An error occurred, please try again.
Interactive Control Panel for USB-6501, USB-6008, and USB-6009 using DAQmx Base
The latter is available on the CD shipped with the device, and ni daq 6009 alternatively be downloaded for free from http: Special USB features are described separately. The Analog Input dialog window in the Test Panels window.
By clicking the Finish button in the window shown in Figure 15, the window shown in Figure 16 is opened where you can select the physical channels for your task. We may then repeat this procedure for other channels. There are ni daq 6009 items available.
National Instruments NI USB 6009 Low-cost Multifunction DAQ
This seller offers ni daq 6009 discounts on combined purchases for eligible dxq. Learn More – opens in a new window or tab Any international shipping is paid in part to Pitney Bowes Inc.
Window where you can enter the name of the task. Input high is between 2.
Ni daq 6009 test procedure, which is denoted loopbackis to connect the AI0 channel to the AO0 channel. This item will ship to United Statesbut the seller has not specified shipping options.
Counting on falling edge. Note how the task value the magenta coloured line is propagated through the relevant functions.
vaq The wizard has finished the installation of the driver software, and the USB is ready for use, after you click Finish Norwegian: Via the Polymorphic VI Selector you can select the type of data the function will write. Ni daq 6009, here I assume that you select Cancel. Please enter 5 or 9 numbers for the ZIP Code.
USB – National Instruments
The New Data Acquisition Device dialog window. Learn More – opens in a new window ni daq 6009 tab Returns: You can ddaq a task for analog output in the same way, that is, ni daq 6009 selecting Create Task in the right-click menu of the USB device as it appears under Devices and Interfacessee Figure 8.
Ni daq 6009 Create Task in the menu opens a window where you can select the measurement type, see Figure The configuration can be saved using the Save button, see Figure Here is a video bi shows the steps: The item you’ve selected was not added to your cart. Most of these are described in the following. The following relevant documents can be found at http: Assuming that MAX recognizes the device, you can now run a self-test of it by selecting Self-Test in the menu.
Shipping cost cannot be calculated. Learn more – opens in a new window or tab. The Voltage Output dialog window in the Test Panels window. A local virtual channel exists only within it’s ni daq 6009 task, while a global virtual channel may be used in several tasks. Note that information about an item in the window is shown in the sub-window down right see Figure